If this is your first visit, be sure to
check out the FAQ by clicking the
link above. You may have to register
before you can post: click the register link above to proceed. To start viewing messages,
select the forum that you want to visit from the selection below.
1 of 4<>
Forum Page Loading Response
We have received feedback regarding recent issues with page loading response time. We believe the slow response issue stems from the analytics we added with the new Forum. We are in the process of reconfiguring those elements and, after an approximate 24-hour time period for the reconfiguration to run, we anticipate a return to normal page load times. Thank you for your patience!
2 of 4<>
Hello, Forno Bravo Community Forum members.
As many may have noticed already, the Community Forum site was briefly down today. While working on scheduled routine maintenance, we encountered an error when trying to add a software update. As the site is now back up and running, some of you may notice that some recent data has been misplaced from November 10th up to today. We are currently working on resolving the issue. The forum has full operational capabilities and we encourage all forum members to continue actively posting in the threads.
We apologize for any inconvenience that this issue may have caused you. The Forno Bravo family values each and every member of our community. If you have any issues or concerns, please feel free to let us know on our issues thread here:
The Forno Bravo team has heard the feedback in regards to the community forum. We wanted to take the time to re-enforce our commitment to a fully engaged Forum with professional moderation.
Our top priority as a company is to fix all forum errors and issues that you are experiencing. As we are swiftly working on these problems, we want to say that we highly value the Forum Bravo Community Forum and every single community forum member.
We have set up this thread so that every member can address any concerns, issues and questions about the forum. Please feel free to ask whatever you would like in regards to the forum; let us know what issues you are experiencing so we can work on resolving them as fast as possible. However, we stress that we would like constructive engagement, so please be specific about the issue you are experiencing.
Thank you for all of your patience and continued support.
For many of you who bought a modular oven, you may have asked how we put the domes together when we build them. For those of you considering one of our ovens, we shot a video to make your install easier.
I have been happy with the performance of the Cen tech laser, but my expectations are low anyway. I only used it to get a general feel of the temps. So much of cooking is done by experience, feel, and on the fly adjustments that the thermometer becomes obsolete fairly quickly. I would get one just for the initial phase of learning to cook and not worry if it is a few degrees off. Search the forum for thermocouple and you will see a lot of the same thing. Some people put in a lot of thermocouples but end up only 1 or 2 ( if that).
As a quick test of my HF Cen Tech infrared thermometer I just boiled a pan of water. As we all know water boils at 212 F at sea level and I live and the test conducted at an elevation just under 200 ft. I had about one inch of boiling water in the pan and after I had a good boil, I turned off the heat and the instant the water surface calmed I took it's temp from a distance of approximately 8 inches. The test was repeated five times (bringing the water back to a boil each time) and the results were: 213.8; 208.6; 210.1; 210.5; and 211.4. The reason (for those curious) I allowed the water surface to calm (bubbles popping on the surface... this took but just a moment) was I wanted to read the temp of the water and not have the reading influenced by the water vapor/steam from the exploding bubbles which is at a higher temperature.
It would be interesting if someone with another brand of infrared thermometer conducted the same test and reported their results.
It is also to be noted that most measuring instruments have a "sweet spot" or range where they are most accurate and the accuracy decreasing outside this range. However, since we would be measuring the same thing in the same range using the same type of measuring device perhaps this is of less importance.